The Very Sharp Cut Cyclone, BX-1020, and the latest BAM User's Manuals are now available to ARB staff
The objective of the Air Monitoring Web Manual is to provide standard procedures for maintaining and operating air monitoring stations and detailed instructions for acceptance testing, maintaining, troubleshooting and calibrating specific analyzers or support equipment.
The accuracy and/or validity of data obtained from any analyzer depends upon analyzer performance and operator proficiency. Deviations from the recommended procedures set forth in this manual, as well as in the Manufacturer's Instruction Manual, may result in the collection of invalid data.
Therefore, the station operator must become familiar with the information contained in this manual as well as each Manufacturer's Instruction Manual in order to achieve a minimum level of competence and acceptable level of data quality.
Note: To print an appendix, first download the file to your computer, then open the document and choose print.
|Number & Type||Document Name||Date Posted||File Size|
|400 sop||Met One Instruments Beta Attenuation Mass Monitor (BAM-1020)||03/28/07||1.02 MB|
|401 sop||Met One Instruments Speciation Air Sampling System||03/28/07||404 KB|
|402 sop AI||Andersen RAAS2.5-300 Sequential Air Sampler||03/28/07||383 KB|
|403 sop AJ||Rupprecht & Patashnick Co., Inc. Partisol-FRM Model 2000 PM-2.5 Air Sampler||03/28/07||192 KB|
|404 sop||Rupprecht & Patashnick Co., Inc. Partisol-Plus Model 2025 Sequential Air Sampler||03/28/07||640 KB|
|406 sop||BGI PQ100 Air Sampler||03/28/07||604 KB|
|407 sop||Magee Scientific Aethalometer||07/07/09||205 KB|
|408 sop||Volumetric Flow Control PM10 Samplers||12/31/10||948 KB|
|409 sop Z||Rupprecht & Patashnick Series 1400a TEOM PM10 Monitor||03/28/07||1.09 MB|
|409 sop||TSI Model 3781 Water-Based Condensation Particle Counter||07/07/09||806 KB|
|410 sop E||High Volume Sampler||03/28/07||1.54 MB|
|411 sop||Met One Instruments E-BAM, Portable Beta Attenuation Mass Monitor||11/17/14||712 KB|